Please use this identifier to cite or link to this item:
|Title:||Sterilisation of tonometers and gonioscopes.|
Honavar, S G
|Citation:||Sood D, Honavar SG. Sterilisation of tonometers and gonioscopes. Indian Journal of Ophthalmology. 1998 Jun; 46(2): 113-6|
|Abstract:||Precautions to prevent spread of infection through tonometers and gonioscopes are described in this article. Tonometers and gonioscopes should not be used in the presence of clinically manifest conjunctivitis and keratitis. The Schiotz tonometer should be dipped in a 1:1000 merthiolate solution, and rinsed in saline/distilled water prior to use. The Goldmann applanation prism tip can be wiped with gauze soaked in 70% isopropyl alcohol and then dried before use. Gonioscopes should be cleaned in running water, wiped with gauze soaked in 70% isopropyl alcohol, and then dried before use. Koeppes and goniotomy lenses can be sterilized with ethylene oxide, prior to use in surgery.|
|Appears in Collections:||Indian Journal of Ophthalmology|
Files in This Item:
There are no files associated with this item.
Items in DSpace are protected by copyright, with all rights reserved, unless otherwise indicated.